**Note: One top and one bottom E-chip are required for every experiment. The Electrochemistry top E-chips have a 500 nm spacer built in for minimum liquid thickness of 500 nm.
Linear Electrode E-chips
The linear electrode configuration allows for flexibility in electrode location and provides you with an option to have two electrodes within the viewing area, allowing an environment where you can see the anode and cathode simultaneously in the TEM. For experiments where two electrodes are desired in the viewing area, ECT-15WR’s electrodes are all 5 μm wide, two of them are located 30 μm apart from each other on the silicon nitride viewing window and the third is 200 μm off the window. For experiments where one electrode is desired in the viewing area, ECT-15WO’s electrodes are all 10 μm wide with one located in the center of the viewing window and the others located 75 μm and 200 μm away, respectively.
All configurations have platinum counter and reference electrode.
A quantity of 1 of any part number indicates 1 box of E-chips. Each box contains 10 E-chips.
TIP: The small E-chip with the 300μm x 90μm window is recommended for broadest viewing area of the working electrode; the small E-chips with the 550μm x 50μm window are recommended for maximizing resolution.
All feature dimensions listed should be considered nominal. Variations in dimensions can occur.